Zero Setting: Press
Change Measuring Standard: Press
* Technical Data
Measuring Range: 0-6”/150mm, 0-8”/200mm,0-12”/300mm
Measuring system: Capacitive
Display: LCD (10.0 mm high)
Maximum Measuring Speed: 120” per second
Battery: 3V, type CR2032 Lithium
Battery life: Approx. 1 year
Operational Temperature Range: 0℃ to+ 40℃
Maximum Relative Humidity: 80%
Once Display beings flashing:
Slide off the battery cover and rubber gasket, and then remove the battery by gently tapping the instrument in your hand. (Never try to face or pry the battery out). Insert the new battery with the positive pole”+” facing upwards and replace the gasket and cover.
Please dispose of used batteries at a proper collection center.
Clean the caliper with a soft cloth DO NOT use any type of solvent. DO NOT immerse in liquid.
In case of water/coolant splash:
The electronic unit for this instrument is water resistant. In case of liquid splash, the display may possibly fail. Wipe off the liquid, be sure the liquid between the slider and beam is completely dried. Then rest the instrument.
* Troubleshooting Chart
Digits do not change or count correctly
Remove the battery for 30 seconds then reinstall.
Check battery contacts or replace battery.
RESET: In order to RESET the instrument, remove the battery, wait 30 second, replace the battery and turn the instrument on.
Precautions: Although a top quality product, certain precautions are required for any electronic instrument:
Avoid exposure to all liquids and excessive humidity.
Avoid exposure to electromagnetic fields.
Do not expose the instrument to direct sunlight.
Do not attempt to disassemble the caliper for extended periods of time.
The purpose of Slider Kit is to gather common slideshow-like jQuery functionalities (such as news sliders, photos galleries/sliders, carousels, tabs menus)
When tracking mouse movement, you usually need to know the actual position of the mouse pointer. The event object that is passed to the handler contains some information about the mouse coordinates.